KiWAVE™ is a trademark of Keithley Instruments, Inc. All Keithley Instruments product names are .. Model Arbitrary Waveform Generator User’s Manual. Items are supplied with manuals, accessories and typically a full no-quibble 2 The exceptional signal quality of the Model is a result of its high resolution, . Refer to the Keithley Model Arbitrary Waveform Generator User’s Manual if it is not clear how to specify the waveforms. The figure below shows an.
|Country:||Turks & Caicos Islands|
|Published (Last):||26 November 2013|
|PDF File Size:||13.88 Mb|
|ePub File Size:||8.37 Mb|
|Price:||Free* [*Free Regsitration Required]|
IP information displayed on this page can be modified. Remote Interface Operation You must have one of the following items installed on your computer: Table of Contents Add to my manuals Add. Software Operation “Model Name” is not alphabetic, the system will pick the first and last letter of the company name for the header, plus “Model Name.
Execution errors Error code Description The maximum output frequency for a pulse Settings conflict; frequency waveform is 10 MHz. If Secure AWG after calibrated was checked, the manual adjustment function will be locked by the secure code after calibration is completed.
KEITHLEY USER MANUAL Pdf Download.
Verification Procedures Recommended adjustment Whatever verification interval you select, Keithley Instruments recommends that complete readjustment should always be performed at the verification interval. The default source is internal with a default frequency of 10 Hz the acceptable range is 2 mHz to 20 kHz. Any unauthorized reproduction, photocopy, or use of the information herein, in whole or in part, without the prior written approval of Keithley Instruments is strictly prohibited.
Rear Panel Description Section 3: Setting Dc Offset Voltage Section 4: Page 5 Any unauthorized reproduction, photocopy, or use the information herein, in whole keithleh in part, without the prior written approval of Keithley Instruments, Inc. Verification procedures Verification tests Use the performance verification tests to verify the measurement performance of the instrument.
Page 98 Appendix A: Service Form Model No.
Sears think[box] | Case School of Engineering
When you change function changed for pulse function from one allowing higher frequency to pulse waveform, the frequency is adjusted to 10 MHz. Introduction Handle adjustment You can adjust the carrying handle to suit your needs. The default carrier frequency is 1 keituley for all waveforms.
An error message will display as shown below, during an calibration error. Also See for User manual – pages.
The default internal modulating waveform is a sine waveform, but you can also select from the following: Checking 0 Db Range Flatness Section 1: Also See for Adjustment manual – 25 pages.
All Keithley Instruments product names are trademarks or registered trademarks of Keithley Instruments, Inc.
Keithley 3390 User Manual
You NOTE can disable the voltage auto ranging function refer to ranging to Setting voltage auto prevent output disruption. Waveform Output Operations The default source is internal with a default frequency of Hz the acceptable range is 2 mHz to 20 kHz.
The default internal modulating waveform is a sine waveform, but you can also select any of the following: C-1 Application Programs About application programs Set the to each output described in the following table and measure the output amplitude with the power meter the relative measurement in dB.
Press the Utility key, and then press the soft ,anual under System to access the system menu. Press the soft key under Type; then Controlling the output signal Signal from the Output connector kanual controlled by the Output key.
Waveform Output Operations Figure Page of 25 Go.
Service Form Model No. Waveform Output Operations Burst phase Burst phase defines the starting phase of a burst.
To print the manual completely, please, download it. Pulse Width Modulation Waveform ns to s. SCPI command summary Query the phase deviation. Pattern Output Operation Section 5: