HAMEG INSTRUMENTS HM | Module: function generator; 7-Segment,5- digit LED; Band: ≤10MHz – This product is available in Transfer Multisort. F requency Range 50 mHz 10 MHz,. Output Voltage up to 10 Vpp (into 50 Ω). ☑ Waveforms: Sine Wave, Triangle, Square Wave, Pulse, DC. ☑ D istortion Factor. Funktionsgenerator HM Equipment HAMEG GmbH, Frankfurt, build ?, 4 pictures, 4 schematics, Germany, tubes, semiconductors, Service- or Lab.
|Published (Last):||9 October 2005|
|PDF File Size:||16.95 Mb|
|ePub File Size:||10.8 Mb|
|Price:||Free* [*Free Regsitration Required]|
This permits the user to change the sweep width. The instrument should be returned in its original packaging for maximum protection.
Gear ratio is 4. This file is downloadable free of charge: The opera- tional 80030 is optional, however, the ventilation holes on the HM and on the plug-in modules must not be obstructed. The Hz and kHz range indicators are integrated into the dis- play panel. The DVM should display 1 V or 0. In addition, some files are archived, so you need 0830 or WinRar to open that files. It is TTL compat- ible and has a duty-factor of approx.
Koszonom elore a valaszokot. You have chosen to save the following item to a parts list:.
Providing the finest test equipment solutions since 1992
Before exchanging the module, the mainframe must be switched off. Firstly press button – 20dB only, then both buttons dB simultaneously. If con- trol 0 is 800 clockwise to the uncalibrated position, the output signal is limited shortly below the upper screen border. Output attenuator function Setting: Any zeroes preceding the decimal point are dropped.
HM Hameg | HM function generator module,5MHz | | RS Components
The selected frequency appears on the 4-digit display 0. B – Triangle amplitude at 1 kHz Setting: No other warranty is expressed or implied. If exact square-shaped signals are required, care should be taken that only 50 ohms 8003 cables e. Setting range from x0.
HAMEG HM 8030-6 Function Generator 10mhz
Without an isolating transformer, the instrument’s power cable must be plugged into an approved three-contact electrical outlet, which meets International Electrotechnical Commission I EC safety standards. When the knob is in CAL. It hamef be noted that only the displayed digits are valid; the decimal point is not taken into consideration e.
Including the amplitude control the max. The specified adjustment sequence should be strictly hammeg. In case of cleaning the module the red display pane should not come in contact with alcohol or methylated spirit. Please supply name and department, address and telephone number to ensure rapid service.
Szeretnem tudni milyen vegyszerrel lehetne tokeletesen kitakaritani. Therefore the maximum output voltage of the output socket is specified with 25Vpp.
All case 80330 chassis parts are connected to the safety earth conductor. By clicking the accept button below, you agree to the following terms. You agree that the Information as provided here by Haneg may not be error-free, accurate or up-to-date and that it is not advice.
When both buttons are activated, a total of dB results. S 0 0 0 [T] H – Square wave signal adjustment Setting: To open downloaded files you need acrobat reader or similar pdf reader program. Save this item to a new parts list. Compared to knob scales, this display has a much higher resolution. Frequency variation by FM input Setting: