The ELx™ is a compact, robust microplate reader ideally suited for applications within the clinical and life science research laboratory. When interfaced with. BioTek – ELxAbsorbance Microplate Reader, The ELx™ is designed for applications within the clinical, biotechnology research and pharmaceutical. BioTek ELx – Operator’s BioTek – ELx by BioTek. BioTek – ELx Download PDF. Product Details · Forums · Documents; Parts; Videos.
|Published (Last):||3 May 2005|
|PDF File Size:||4.57 Mb|
|ePub File Size:||3.44 Mb|
|Price:||Free* [*Free Regsitration Required]|
Discard the used gloves and cloths using a Biohazard trash bag and an approved Biohazard container.
Bio-tek ELx800 Operator’s Manual
Select YES to indicate that one or more standard replicates or groups should be temporarily excluded from curve-fit calculations. The well plate type is preset for the map to have samples. Printer timed out The time allotted manial the instrument to make a valid connection to a printer has expired. Chapter 6 Troubleshooting and Error Codes This appendix describes potential error codes that may be displayed mnual the ELx, and suggests possible solutions for these problems.
Page Error Codes General Errors Continued Errors 0D00 through indicate that the reader has failed one or more of a number of self-tests. Page 46 – Figure 4: The maximum number of control groups is 8. It is recommended that each laboratory determine its own normal range, based on the population techniques and equipment employed. Table of Contents Add to my manuals Add. To capture and manipulate the raw data using well geometry, you must use KCjunior or KC4.
Page Chapter 5: Print the raw data by selecting Plate Print Results. Read Plate dialog Choose New Protocol to define the protocol. Data new Plate Dialog Figure Press Options to cycle through the assay numbers and names, or use numeric keys to enter an assay number.
The ELx has a pin serial RS port located on the rear panel of the instrument. Examples of mapping directions are shown on the next page. Hooking the shipping straps around the shaft Bio-Tek Instruments, Inc.
Use the slotted screwdriver to remove the four screws from the top cover mounting Figure 3 and shipping straps from the bottom of the instrument. Do not soak the cloth.
BIO-TEK ELX OPERATOR’S MANUAL Pdf Download.
If the assay contains a closed variable i. This packaging should protect the instrument from damage during shipping. The plate will be read and then the raw data results will display in KCjunior. Removed previous Appendix C, Error Codes. Calculate Mean Optical Densities for assays run in duplicate only For each standard, control and sample calculate the mean OD of the duplicate readings.
Do not allow the cleaning solution to run into the interior of the instrument.
For the latest list of compatible printers, consult the Bio-Tek Web site www. Selecting 6-,or well geometries enables data reduction capabilities. For comparison in this test, the following repeatability spec is applied, using Normal mode, well microplate: Connectors for Printer Parall Subtract the difference from all ODs maunal the plate. Page – Entering the Absorbance Plate Specificat Do not turn on the instrument until the carrier shipping block has been removed.
Precautions Preface Precautions The following precautions are provided to help avoid damage to the instrument: Ex800 a PBS solution using the Sigma powder. Set up the ELx according to the instructions in Chapter 3, Installation. If the four corner wells are within the accuracy range, the reader is in alignment. Forums Documents and Manuals.
Registration Card Once the instrument has been set up and is running successfully, please take a moment to fill out and mail the postage-paid Warranty Registration card.